Journal of Contemporary Brachytherapy

J Contemp Brachytherapy. 11(6): 573-578

A new implant device to prevent edema-associated underdosage in high-dose-rate interstitial brachytherapy of mobile tongue cancer

Ken Yoshida, MD, PhD¹, Tadayuki Kotsuma, MD², Hironori Akiyama, DDS³, Hideya Yamazaki, MD⁴, Tadashi Takenaka, RTT⁴, Koji Masui, MD⁴, Yutaka Tsujimoto, RTT⁵, Naoya Murakami, MD⁶, Yasuo Uesugi, MD⁷, Taiju Shimbo, MD¹, Nobuhiko Yoshikawa, MD¹, Hiroto Yoshioka, MD¹, Mio Nakata, MD¹, Takumi Arika, DDS⁸, Yuji Takaoka, MD², Eiichi Tanaka, MD², Nikolaos Tselis, MD⁹

1. Department of Radiation Oncology, Osaka Medical College, Takatsuki, Osaka, Japan,

2. Department of Radiation Oncology, National Hospital Organization Osaka National Hospital, Osaka, Japan,

3. Department of Oral Radiology, Osaka Dental University, Osaka, Japan,

4. Department of Radiology, Kyoto Prefectural University of Medicine, Kyoto, Japan,

5. Department of Radiology, National Hospital Organization Osaka National Hospital, Osaka, Japan,

6. Department of Radiation Oncology, National Cancer Center Hospital, Japan,

7. Department of Rehabilitation Sciences, Faculty of Allied Health Sciences, Kansai University of Welfare Sciences, Japan,

8. Department of Oral Surgery, National Hospital Organization Osaka National Hospital, Osaka, Japan,

9. Department of Radiation Oncology, Johann Wolfgang Goethe University Frankfurt, Germany

Address for correspondence: Ken Yoshida, MD, PhD, Department of Radiation Oncology, Osaka Medical College, 2-7, Daigaku-machi, Takatsuki, Osaka 569-8686, Japan, e-mail: rad113@osaka-med.ac.jp

Copyright © 2019 Termedia

DOI: 10.5114/jcb.2019.91225

Published online: 25 December 2019

Published in print: December 2019

Abstract

Purpose

Tongue edema is a potential cause of treatment target underdosage in high-dose-rate interstitial brachytherapy (HDR-ISBT) of mobile tongue cancer. To prevent such edema-associated alteration of dosimetry, we developed a special silicon device. In this report we communicate our initial experience with two mobile tongue cancer patients whom we treated using this new device.

Material and methods

The device consists of silicone tubes with a fixed width and scalable length depending on tongue size. These tubes are lined and fixed like a palisade, allowing the device to be used also as a template. The device is placed next to the lateral border of the tongue and on the floor of the mouth. In addition, a vinyl template can be placed on the dorsal tongue surface with both devices combined for implantation guidance. Between June and August 2012, two patients with locally confined tongue cancer were treated.

Results

Between June and August 2012, two mobile tongue cancer patients classified as cT2N0M0 were treated with HDR-ISBT using the silicone device. They underwent ISBT as monotherapy with fractional doses of 6.0 Gy up to a total physical dose of 54.0 Gy. The D₉₀ (CTV) values of both patients were 6.3 Gy and 6.6 Gy and the D2cc (mandible) values were 3.4 Gy and 2.6 Gy, respectively. At present, both patients remain without local disease recurrence at 60 and 56 months after ISBT, respectively.

Conclusions

The described silicone device has the potential to prevent underdosage to the treatment target related to tongue edema. It has been shown to be safe and easy to implement.

Purpose

High-dose-rate interstitial brachytherapy (HDR-ISBT) [1,2,3,4] is an established radiotherapy (RT) modality for the treatment of tongue cancer. Computed tomography (CT)-based dosimetry will clarify in HDR-ISBT if there is any alteration of the geometrical relationship between interstitial catheter positions and the clinical target volume (CTV), which can occur during fractionated treatments. In analogy to this scenario, edema and applicator displacement are potential causes of under- or overdosage of the CTV, as well as the organs at risk (OAR), in prostate and gynecologic HDR-ISBT [5,6,7,8,9]. Similarly, tongue edema can impair target coverage in temporary interstitial tongue RT [10,11,12,13]. To prevent such alteration of CTV coverage, we introduced a new device which aims to avoid an edema-associated shift of tongue tissue (mucosal surface) beyond the prescription isodose line. In this report we communicate our initial clinical experience with two mobile tongue cancer patients whom we treated with HDR-ISBT using the new silicone device in order to prevent treatment-related tongue edema.

Material and methods

Silicone device

Our treatment device consists of silicone tubes with a fixed width (5 mm outer diameter and 3 mm internal diameter) and a scalable length of 10-15 mm depending on tongue size. These tubes are lined and fixed at both ends with two pieces of string in a transverse direction like a palisade, which allows the device to be used as a template (Figure 1A). In the oral cavity the device is positioned next to the lateral border of the tongue and on the floor of the mouth. This also enables the lateral border of the tongue to be irradiated adequately by implanting catheters into every three silicone tubes, ensuring an applicator-applicator interval of about 15 mm. In addition, a vinyl template can be placed on the dorsal tongue surface (Figure 1B) with both devices combined for the purpose of implantation guidance (Figure 1C).

Fig. 1View larger versionFig. 1. A) Photograph of a silicone device. The width of the silicone tube was 5 mm outer diameter and 3 mm internal diameter, and its length was 10-15 mm depending on tongue size. These tubes were lined and fixed at both ends with two pieces of string in a transverse direction like a palisade. This device was positioned next to the lateral border surface of the tongue and on the floor of the mouth. We could use the device as a template. B) Photograph of a vinyl template. It was placed on the dorsal tongue surface. C) Photograph of combination with a silicone device and vinyl template. Flexible treatment applicator tubes (Flexible implant tube, 6 F/1.9 mm external diameter, single leader; Elekta AB, Stockholm, Sweden) were inserted

Implantation

Clinical implementation of the new silicon device is part of our “classical” tongue implantation procedure. In short, open-end catheters (Bevel needle; Elekta AB, Stockholm, Sweden) are implanted from the submandibular region into the oral cavity. To treat adequately the lateral side of the tongue, the catheters are implanted into the floor of the mouth close to the lateral border of the tongue (Figure 2A). The silicone device is at first placed on the floor of the mouth and the open-end catheters are penetrated from the floor surface into the silicone device and finally into the holes of the vinyl template. After completion of implantation, all open-end catheters are replaced by flexible applicator tubes (Flexible implant tube, 6 F/1.9 mm external diameter, single leader; Elekta AB) and these tubes fix silicon devices in a longitudinal direction. For medial parts of the tongue, open-end catheters are implanted through the dorsal surface of the tongue into the vinyl template and replaced by flexible applicator tubes (Figure 2B). Concerning the movement of flexible applicator tubes inside the silicon tubes, the applicators are fixed longitudinally because flexible applicator tubes penetrate tissue from the submandibular region to the mouth floor and the tubes are fixed by the button and vinyl template at the dorsum of the tongue. Laterally the maximum distance of this movement is 0.55 mm because the external diameter of the flexible applicator tubes is 1.9 mm and the internal diameter of the silicon tubes is 3 mm. 0.55 mm is a very small number and we consider that it is not necessary to take account into the movement of flexible applicator tubes inside silicon tubes for treatment planning. Before implantation, titanium markers are implanted at the edge of the gross tumor to aid CTV delineation for image-based treatment planning.

Fig. 2View larger versionFig. 2. A) Photograph of applicator implantation for case 1. Metallic open-end needle (Bevel needle; Elekta AB, Stockholm, Sweden) was implanted from submandibular region to oral cavity. B) Photograph of the same patient as Fig. 2A. Flexible treatment applicator tubes (Flexible implant tube, 6 F/1.9 mm external diameter, single leader; Elekta AB, Stockholm, Sweden) were implanted into the silicone device and vinyl template. We also inserted a silicone block between the lateral border of the tongue and gingiva

Planning and treatment

Our three-dimensional (3D) treatment planning method has been described elsewhere [14,15]. In short, treatment planning is CT-based. We draw gross tumor volume (GTV), CTV and mandible as an OAR. CTV results from GTV plus at least a 5.0 mm margin and incorporates clinical findings from inspection, palpation, and intraoperative intra-oral ultrasound as well as titanium marker orientation. Planning is based on a modified Paris system approach with anatomy-oriented dose optimization (Figure 3A) [16]. Treatment plan evaluation is based on dose-volume histogram (DVH) analysis under consideration of the dose that covers 90% [D₉₀ (CTV)] as well as 100% of the CTV [D₁₀₀ (CTV)]. Our dosimetric goal is D₁₀₀ (CTV) ≥ prescription dose (PD); however, excessive doses to OARs (e.g. mandible) or a hyperdose sleeve > 10 mm allow for D₁₀₀ (CTV) < PD. In such cases, our dosimetric goal is D₉₀ (CTV) > PD. For DVH-based OARs evaluation, we use the minimum dose received by the maximally irradiated 2 cc (D2cc) of the mandible. For this, we use lead blocks as spacers and shielding devices to the gingival mucosa and mandible for protection purposes [17]. For CT planning, however, we use silicone blocks instead of lead blocks to prevent metal artifacts (Figure 2B).

Fig. 3View larger versionFig. 3. A) An isodose shape for case 1. The gloss tumor volume (black dotted line) and clinical target volume (white line) were almost covered by the 100% prescribed isodose line (red dotted line). The silicone device was clearly visualized just laterally of the tongue. The silicone block (Block) was also visualized without any artifact. B) An isodose shape for case 2. The gloss tumor volume (black dotted line) and clinical target volume (white line) could be covered by the 100% prescribed isodose line (red dotted line). The silicone device was clearly visualized just laterally of the tongue. The silicone block (Block) was also visualized without any artifact

Our ISBT protocol consists in the monotherapy setting of fractional HDR doses of 6.0 Gy up to a total PD of 54.0 Gy in nine fractions over 7 days. Two fractions were administered per day. The time interval between fractions was > 6 h. The implantation and first treatment were done on Wednesday, and the final treatment and the extraction of the flexible applicator tubes on the next Tuesday. The patients were not treated on Saturday and Sunday. This generates a biological effective dose (α/β = 10) of 86.4 Gy, and equals an equivalent total dose in 2-Gy fractions (EQD₂) of 72 Gy. All treatments are performed using an iridium-192 (¹⁹²Ir) HDR-afterloading system (microSelectron-HDR, Elekta AB, Stockholm, Sweden) with an apparent initial source activity of approximately 370 GBq. During the treatment period the implant devices were always put along the lateral border of the tongue.

Clinical cases

Between June and August 2012, two mobile tongue cancer patients were treated by HDR-ISBT using the new silicone device at National Hospital Organization Osaka National Hospital.

Case 1

An 84-year-old male patient with heavy smoking history presented in June 2012 at our department with squamous cell carcinoma on the right lateral border of the tongue. Tumor stage was cT2N0M0 according to the 2007 UICC classification system. After rejecting glossectomy as well as definitive external-beam radiotherapy (EBRT), he was offered HDR-ISBT as monotherapy (Figure 3A). Treatment consisted of 54.0 Gy delivered in nine fractional HDR doses of 6.0 Gy over 7 days. The volumetrically calculated GTV and CTV were 5.3 cc and 17 cc, respectively. The generated D₉₀ (GTV) was 6.6 Gy and D₁₀₀ (GTV) 6.0 Gy. The respective D₉₀ (CTV) was 6.3 Gy and D₁₀₀ (CTV) 5.0 Gy with D2cc (mandible) at 3.4 Gy.

At 60 months after ISBT the patient remains free from disease without local or systemic progression. However, at 32 months after treatment he presented osteoradionecrosis of the mandible which requires pain management with oxycodone hydrochloride but no intervention or hospitalization. Against medical advice, he did not quit smoking despite experiencing symptomatic osteonecrosis [18].

Case 2

A 38-year-old female patient presented in August 2012 at our department with squamous cell carcinoma on the left lateral border of the tongue. Tumor stage was cT2N0M0 according to the 2007 UICC classification system. After rejecting glossectomy, she was offered HDR-ISBT as monotherapy (Figure 3B). Treatment consisted of 54.0 Gy delivered in nine fractional HDR doses of 6.0 Gy over 7 days. The volumetrically calculated GTV and CTV were 1.3 cc and 6.1 cc, respectively. The generated D₉₀ (GTV) was 6.6 Gy and D₁₀₀ (GTV) 6.3 Gy. The respective D₉₀ (CTV) was 6.6 Gy and D₁₀₀ (CTV) 6.1 Gy with D2cc (mandible) at 2.6 Gy.

At 56 months after ISBT the patient remains free from local disease recurrence. However, 3 months after treatment she was diagnosed with cervical lymph node metastasis and received radical neck dissection with postoperative EBRT up to 60 Gy in 30 fractions. At present, she remains free from local or further systemic progression. No mandible complication was observed.

Discussion

Image-based HDR-ISBT has proven to be an effective modality in the treatment of tongue cancer by escalating the biologically effective dose to the treatment target whilst ameliorating conformity [1,2,3,4]. GEC-ESTRO ACROP recommendations [19] reported that brachytherapy alone has been nowadays replaced by surgery in the treatment of most T1 and T2 tumors due to advances in surgical and anesthetic procedures that have been proven safe and effective even in frail patients; however, no randomized trial has ever compared surgery versus brachytherapy in primary tumors. In addition, surgery provides a complete pathological documentation of the extent of disease that allows an individualized adjuvant treatment plan for subsequent radiotherapy. However, brachytherapy alone remains an acceptable mode of treatment in intact T1 and small T2 tumors with low risk of lymph node involvement that meets at least one of the 5 following criteria. Patient decision is one of the criteria, and our cases were applicable because they rejected glossectomy as well as definitive EBRT. 3D treatment planning in HDR enables anatomy-oriented dose optimization while the versatility of intratarget dose modulation inherent to ISBT can be controlled and directed to deliver higher doses to gross disease or to selectively reduce the dose to OARs [20]. However, the rapid dose fall-off in HDR can also become a cause of underdosage to the treatment target if tongue edema occurs. Schultze et al. [10] investigated the incidence and extent of tongue edema in 51 patients with predominantly base of tongue cancer, reporting an applicator-induced edematous deviation of tongue width of median 6.0 mm as measured by MRI. In our own experience, we also encountered tongue edema after interstitial tongue implantation which resulted in treatment target underdosage and eventually local disease recurrence.

In our previous study, the coverage of the lateral border of the tongue was 100% of the PD at first but it was decreased to 70% PD because part of the mucosal surface protruded between two flexible applicator tubes by tongue edema if we did not use the new implant device [13]. The D₉₀ (CTV) values were 6.3 Gy (105% PD) and 6.6 Gy (110% PD) in this study. In our previous study, the mean D₉₀ (CTV) was 112.4 ±6.1% of the PD for the image-based plan [15]. And so, the DVH data of this study were reasonable if no displacement occurred.

In order to avoid such alterations in dosimetry, especially during fractionated treatments, we developed the described silicone device which is characterized by two significant merits. Firstly, the device can limit the anatomic extension of treatment-associated tongue edema within its own borders. It is clearly identifiable in CT imaging, therefore supporting target delineation and reproducible treatment planning (Figure 3A,B). Ideally it may be better to take CT images for accurate evaluation of edema and make a new plan at each fractionated treatment, but spine position is hard and tiring for patients because of difficulty swallowing due to irritation of the applicator tubes. Therefore we took CT images and made a plan only once. Edema can change the shape and volume of the tongue in every direction. Our new devices prevent edema in lateral, anterior and dorsal sides. For the medial and posterior aspect of the tongue we applied flexible applicator tubes to maintain a sufficient safety margin from the GTV. Concerning the ventral side we took into account edema when we decided on dwell positions of the treatment source. Therefore we consider that edema-induced changes of tongue volume for all directions were reflected in our treatment. However, we will verify them one by one. Secondly, the device can serve as a template allowing even inexperienced brachytherapist to perform tongue implants with greater confidence and dosimetrically advantageous.

Notwithstanding, with regard to clinical outcomes the patient in case 1 developed symptomatic osteoradionecrosis. Even though D2cc (mandible) was only 3.4 Gy per fraction and 30.6 Gy in terms of total physical dose, our device might have contributed to avoiding local disease recurrence but did not prevent OAR toxicity. It remains doubtful, however, whether a dose-effect relationship exists for such a relatively low dose or the necrosis was associated with continuation of smoking. Zevallos et al. [18] investigated 86 patients receiving EBRT for laryngopharyngeal cancer and reported that continuation of smoking was strongly associated with the development of osteoradionecrosis (relative risk [RR]: 1.32; 95% confidence interval [CI]: 1.09-1.6, p = 0.03).

We consider the introduced silicon device a meaningful tool which has the potential to keep an interstitial tongue implant three-dimensionally stable and to contribute to OARs sparing due to reproducible dosimetry. Its implementation and general rationale are obvious to the radiation oncologist. However, some aspects need to be emphasized concerning its applicability. First, the silicone device can only be positioned at the floor of the mouth without the option to use it for base of tongue lesions. In addition, it must be individually assembled for various clinical settings. This flexibility certainly extends its safe applicability but might also be an expenditure of time, albeit without relevance when considering its potential merits.

The limitation of this initial experience study is that the number of patients is only two, and it is difficult to draw firm conclusions. The other limitation of this study is the lack of evaluation of edema during treatment. Further studies with a larger number of patients using retrospective comparison of matched cohorts and edema evaluation by CT images at each ISBT are required.

Conclusions

The described silicone device has the potential to prevent underdosage to the CTV potentially related to treatment-associated tongue edema. It has been proven to be safe and easy to implement.

Acknowledgements

This work was supported by JSPS KAKENHI (Grant-in-Aid for Scientific Research (C)) Grant Numbers JP17K10496, JP18K07639, JP17K10488, JP16K11534 and JP15K15461. This work was also supported in part by Japan Agency for Medical Research and Development (JP18ck0106305h002, JP19ck0106305h003) and OMC Internal Research Grant.

We would like to thank Mineo Yoshida, MD, Shunsuke Miyake, RTT, Mari Mikami Ueda, RTT, Toshiro Kajihara, RTT, Katsuhito Kubo, RTT, the other staff of the departments of Radiology, Oral Surgery, Anesthesiology and nursing staff for helping us in many ways during the completion of this study.

Disclosure

The authors report no conflict of interest.

Articles from Journal of Contemporary Brachytherapy are provided here courtesy of Termedia Publishing

PMC Copyright Notice

The articles available from the PMC site are protected by copyright, even though access is free. Copyright is held by the respective authors or publishers who provide these articles to PMC. Users of PMC are responsible for complying with the terms and conditions defined by the copyright holder.

Users should assume that standard copyright protection applies to articles in PMC, unless an article contains an explicit license statement that gives a user additional reuse or redistribution rights. PMC does not allow automated/bulk downloading of articles that have standard copyright protection.

See the copyright notice on the PMC site, https://www.ncbi.nlm.nih.gov/pmc/about/copyright/, for further details and specific exceptions.

References

T InoueT TeshimaS Murayamaet al Phase III trial of high and low dose rate interstitial radiotherapy for early oral tongue cancer. Int J Radiat Oncol Biol Phys 1996; 36: 1201-1204.8985043

T InoueT TeshimaK Yoshidaet al Phase III trial of high-vs. low-dose-rate interstitial radiotherapy for early oral tongue cancer. Int J Radiat Oncol Biol Phys 2001; 51: 171-175.11516867

T TeshimaT InoueH Ikedaet al Phase I/II study of high-dose rate interstitial radiotherapy for head and neck cancer. Strahlenther Onkol 1992; 168: 617-621.1455289

Z Takácsi-NagyC PolgárF Obernaet al Interstitial high-dose-rate brachytherapy in the treatment of base of tongue carcinoma. Strahlenther Onkol 2004; 180: 768-775.15592696

AA MartinezI PatakiG Edmundsonet al Phase II prospective study of the use of conformal high-dose-rate brachytherapy as monotherapy for the treatment of favorable stage prostate cancer: a feasibility report. Int J Radiat Oncol Biol Phys 2001; 49: 61-69.11163498

SJ DamoreAM SyedAA Puthawalaet al Needle displacement during HDR brachytherapy in the treatment of prostate cancer. Int J Radiat Oncol Biol Phys 2000; 46: 1205-1211.10725633

FL CuryM DuclosA Aprikianet al Prostate gland edema after single-fraction high-dose rate brachytherapy before external beam radiation therapy. Brachytherapy 2010; 9: 208-212.20149758

JD KifferWA SchumerCA Mantleet al Impact of oedema on implant geometry and dosimetry for temporary high dose rate brachytherapy of the prostate. Australas Radiol 2003; 47: 172-176.12780447

K YoshidaH YamazakiT Kotsumaet al Treatment results of image-guided high-dose-rate interstitial brachytherapy for pelvic recurrence of uterine cancer. Brachytherapy 2015; 14: 440-448.25858904

J SchultzeR SchneiderB Kimmiget al Interstitial brachytherapy in the head and neck region: when has imaging for treatment planning to be performed? Rontgenpraxis 2002; 54: 232-237.12174447

S NagER CanoDJ Demaneset al The American Brachytherapy Society recommendation for high-dose-rate brachytherapy for head-and-neck carcinoma. Int J Radiat Oncol Biol Phys 2001; 50: 1190-1198.11483328

JJ MazeronJM ArdietC Haie-Mederet al GEC-ESTRO recommendations for brachytherapy for head and neck squamous cell carcinomas. Radiother Oncol 2009; 91: 150-156.19329209

K YoshidaH YamazakiT Kotsumaet al Edema worsens target coverage in high-dose-rate interstitial brachytherapy of mobile tongue cancer: a report of two cases. J Contemp Brachytherapy 2017; 9: 66-70.28344606

H AkiyamaK YoshidaH Yamazakiet al High-dose-rate interstitial brachytherapy for mobile tongue cancer: preliminary results of a dose reduction trial. J Contemp Brachytherapy 2014; 6: 10-14.24790616

K YoshidaT TakenakaH Akiyamaet al Three-dimensional image-based high-dose-rate interstitial brachytherapy for mobile tongue cancer. J Radiat Res 2014; 55: 154-161.23732769

K YoshidaT NoseM Koizumiet al The usefulness of metal markers for CTV-based dose prescription in high-dose-rate interstitial brachytherapy. J Jpn Soc Ther Radiol Oncol 2002; 13: 253-260.

S MurakamiRG VerdonschotN Kakimotoet al Preventing complications from high-dose rate brachytherapy when treating mobile tongue cancer via the application of a modular lead-lined spacer. PLoS One 2016; 11: e0154226.27128434

JP ZevallosMJ MallenCY Lamet al Complications of radiotherapy in laryngopharyngeal cancer: effects of a prospective smoking cessation program. Cancer 2009; 115: 4636-4644.19569250

G KovácsR Martinez-MongeA Budrukkaret al GEC-ESTRO Head & Neck Working Group. GEC-ESTRO ACROP recommendations for head & neck brachytherapy in squamous cell carcinomas: 1st update-Improvement by cross sectional imaging based treatment planning and stepping source technology. Radiother Oncol 2017; 122: 248-254.27889184

H AkiyamaT MajorC Polgáret al Dose-volume analysis of target volume and critical structures in computed tomography image-based multicatheter high-dose-rate interstitial brachytherapy for head and neck cancer. J Contemp Brachytherapy 2017; 9: 553-560.29441100

[Back]

Fig. 1.

A) Photograph of a silicone device. The width of the silicone tube was 5 mm outer diameter and 3 mm internal diameter, and its length was 10-15 mm depending on tongue size. These tubes were lined and fixed at both ends with two pieces of string in a transverse direction like a palisade. This device was positioned next to the lateral border surface of the tongue and on the floor of the mouth. We could use the device as a template. B) Photograph of a vinyl template. It was placed on the dorsal tongue surface. C) Photograph of combination with a silicone device and vinyl template. Flexible treatment applicator tubes (Flexible implant tube, 6 F/1.9 mm external diameter, single leader; Elekta AB, Stockholm, Sweden) were inserted

[Back]

[Back]

Fig. 2.

A) Photograph of applicator implantation for case 1. Metallic open-end needle (Bevel needle; Elekta AB, Stockholm, Sweden) was implanted from submandibular region to oral cavity. B) Photograph of the same patient as Fig. 2A. Flexible treatment applicator tubes (Flexible implant tube, 6 F/1.9 mm external diameter, single leader; Elekta AB, Stockholm, Sweden) were implanted into the silicone device and vinyl template. We also inserted a silicone block between the lateral border of the tongue and gingiva

[Back]

[Back]

Fig. 3.

A) An isodose shape for case 1. The gloss tumor volume (black dotted line) and clinical target volume (white line) were almost covered by the 100% prescribed isodose line (red dotted line). The silicone device was clearly visualized just laterally of the tongue. The silicone block (Block) was also visualized without any artifact. B) An isodose shape for case 2. The gloss tumor volume (black dotted line) and clinical target volume (white line) could be covered by the 100% prescribed isodose line (red dotted line). The silicone device was clearly visualized just laterally of the tongue. The silicone block (Block) was also visualized without any artifact

[Back]