GMS Krankenhaushygiene Interdisziplinar

GMS Krankenhhyg Interdiszip. 2(2): Doc54

Principles and working mechanisms of water-filtered infrared-A (wIRA) in relation to wound healing

Gerd Hoffmann*¹

1. Johann Wolfgang Goethe University Frankfurt/Main, Institute of Sports Sciences, Frankfurt/Main, Germany

*To whom correspondence should be addressed: Gerd Hoffmann, Johann Wolfgang Goethe University Frankfurt/Main, Institute of Sports Sciences, Ginnheimer Landstrasse 39, D-60487 Frankfurt/Main, Germany, Tel+Fax+Q: 0049-6181-62287, E-mail: Hoffmann@em.uni-frankfurt.de

Copyright © 2007 Hoffmann

Published online: 28 December 2007

Abstract

The experience of the pleasant heat of the sun in moderate climatic zones arises from the filtering of the heat radiation of the sun by water vapor in the atmosphere of the earth. The filter effect of water decreases those parts of infrared radiation (most parts of infrared-B and -C and the absorption bands of water within infrared-A), which would cause – by reacting with water molecules in the skin – only an undesired thermal load to the surface of the skin. Technically water-filtered infrared-A (wIRA) is produced in special radiators, whose full spectrum of radiation of a halogen bulb is passed through a cuvette, containing water, which absorbs or decreases the described undesired wavelengths of the infrared radiation. Within infrared the remaining wIRA (within 780-1400 nm) mainly consists of radiation with good penetration properties into tissue and therefore allows – compared to unfiltered heat radiation – a multiple energy transfer into tissue without irritating the skin, similar to the sun’s heat radiation in moderate climatic zones. Typical wIRA radiators emit no ultraviolet (UV) radiation and nearly no infrared-B and -C radiation and the amount of infrared-A radiation in relation to the amount of visible light (380-780 nm) is emphasized.

Water-filtered infrared-A as a special form of heat radiation with a high tissue penetration and with a low thermal load to the skin surface acts both by thermal (related to heat energy transfer) and thermic (temperature depending, with a relevant change of temperature) as well as by non-thermal (without a relevant transfer of heat energy) and non-thermic (not depending on temperature, without a relevant change of temperature) effects. wIRA produces a therapeutically usable field of heat in the tissue and increases tissue temperature, tissue oxygen partial pressure, and tissue perfusion. These three factors are vital for a sufficient tissue supply with energy and oxygen. As wound healing and infection defense (e.g. granulocyte function including their antibacterial oxygen radical formation) depend decisively on a sufficient supply with energy and oxygen, one explanation for the good clinical effect of wIRA on wounds and wound infections can be the improvement of both the energy supply per time (increase of metabolic rate) and the oxygen supply. In addition wIRA has non-thermal and non-thermic effects, which are based on putting direct stimuli on cells and cellular structures.

wIRA can considerably alleviate the pain (with remarkably less need for analgesics) and diminish an elevated wound exudation and inflammation and can show positive immunomodulatory effects. wIRA can advance wound healing or improve an impaired wound healing both in acute and in chronic wounds including infected wounds. Even the normal wound healing process can be improved.

wIRA is contact-free, easily applied, without discomfort to the patient, with absent consumption of material and with a good effect in the depth. The irradiation of the typically uncovered wound is carried out with a wIRA radiator.

What is water-filtered infrared-A (wIRA)? Why wIRA?

The experience of the pleasant heat of the sun in moderate climatic zones arises from the filtering of the heat radiation of the sun by water vapor in the atmosphere of the earth [1], [2], [3], see Figure 1 (Fig. 1). Mankind has developed in the evolution under the influence of this water-filtered heat radiation of the sun [4]. In contrast to this in the desert the sun is stinging and burning, as the water vapor is missing there in the atmosphere of the earth.

The filter effect of water decreases those parts of infrared radiation (most parts of infrared-B and -C and the absorption bands of water within infrared-A), which would cause – by reacting with water molecules in the skin – only a thermal load to the surface of the skin [1], [2], [3], see Figure 1 (Fig. 1).

Technically water-filtered infrared-A (wIRA) is produced in special radiators, whose whole incoherent broad-band radiation of a 3000 Kelvin halogen bulb is passed through a cuvette, containing water, which absorbs or decreases the described undesired wavelengths within infrared [1], [2], [4], see Figure 2 (Fig. 2): an example of the resulting spectrum with visible light (VIS) and water-filtered infrared-A (wIRA) is shown in Figure 3 (Fig. 3).

Within infrared the remaining wIRA (within 780-1400 nm) mainly consists of radiation with good penetration properties into tissue and therefore allows – compared to unfiltered heat radiation of conventional infrared bulbs with large amounts of infrared-B (defined as 1400-3000 nm) and -C (defined as 3000-1,000,000 nm) – a multiple energy transfer into tissue without irritating the skin (high energy transfer with limited temperature increase), similar to sun heat radiation in moderate climatic zones [1], [2].

Based on the water-filtering typical wIRA radiators emit almost no infrared-B and -C radiation and have a decreased irradiance of the absorption bands of water within infrared-A. In contrast to the sun typical wIRA radiators emit no ultraviolet (UV) radiation and the amount of infrared-A radiation in relation to the amount of visible light (380-780 nm) is emphasized (depending on the filtering of the visible light, approximately 75% of the radiation is water-filtered infrared-A), see Figure 4 (Fig. 4).

The water-filtering leads to high penetration properties with a low thermal load to the surface of the skin, see Figure 5 (Fig. 5). This is the basis that wIRA is able to essentially improve even energy-related specific factors of tissue metabolism, especially in regeneration or impaired conditions like wounds. Compared to other infrared radiation sources, there is typically no sense of discomfort or burning during irradiation with wIRA when applying an appropriate irradiance.

Within the spectra of infrared-A and water-filtered infrared-A radiation effects especially of the energy-rich wavelengths near to visible light – approximately 780-1000 nm (800-900 nm [5], [6], [7], 800 nm [8], 820 nm [9], [10], [11], 830 nm [12]) - have been described both in vitro and in vivo, and these wavelengths seem to represent the clinically most important part within infrared-A and wIRA [13], see as well section about non-thermal and non-thermic effects below.

For special purposes, like photodynamic therapy (PDT), the filtering of the visible light can be adapted to special recommendations, see section perspectives in [14].

Working mechanisms of wIRA

wIRA acts both by thermal and thermic effects as well as by non-thermal and non-thermic effects [1], [2].

“Thermal” effects of wIRA are related to a (heat) energy transfer (transfer of heat radiation energy, transfer of infrared radiation energy) e.g. into tissue.

The production of a therapeutically usable field of heat in tissue and the energy-related aspects derived from this belong to the thermal (and thermic, see below) effects of wIRA.

Thermal and thermic effects of wIRA and production of a therapeutically usable field of heat

As an effect related to the transfer of heat radiation energy (“thermal” effect) wIRA produces a therapeutically usable field of heat in tissue [1], [2] by

reaching capillaries near the surface of the skin by the infrared-A radiation (primary warming)

heat distribution by the blood (cooling of tissue areas near the surface of the skin, spreading of the heat into the depth)

increasing capillary bloodflow near the surface of the skin with expansion of the blood flow areas accessible to the radiation and by this augmenting the second mechanism

conduction of heat into the depth

secondary energy release by stimulation of metabolism (increase of metabolism) caused by the increase of temperature (in accordance with the reaction velocity temperature rule a 3°C higher temperature means approximately 30% more speed of reaction and by this more energy provision and release in the tissue)

relatively high primary depth effectiveness of wIRA.

Thermal effects of wIRA – transfer of heat energy, e.g. bringing an amount of heat (heat energy) into tissue – can lead to a temperature increase in the tissue and by this to temperature depending (“thermic”) effects (like the desired sufficient energy production in the tissue, see below). Therefore thermal effects can represent the basis of such thermic effects.

Based on its special properties, wIRA allows a high energy transfer into tissue (with relevant thermal effects) combined with a limited temperature increase in the tissue (limited thermic effects, ideal dosing properties). This is especially of importance concerning safety aspects, see separate section about safety aspects below.

An additional example of a temperature depending effect (thermic effect) of wIRA might be the activating of transient receptor potential channels of the vanilloid (TRP-V) family depending on the increased temperature.

Non-thermal and non-thermic effects of wIRA

Non-thermal and non-thermic effects of wIRA (without a relevant transfer of heat energy and without a relevant change of temperature) are based on putting a stimulus on cells and cellular structures as a specific (direct) radiation effect. Reactions of the cells at infrared radiation – even partly at very small irradiances – are e.g. target oriented growth of surface extensions (plasmodia) [5], influence on the cytochrome c oxidase [9], [15], [16], target oriented growth of neurons [8], stimulation of wound repair [17], [18] as well as cell protective effects of infrared-A [19], [20], [21], [22] (including signalling pathway [20], [21]) and water-filtered infrared-A (wIRA) [23], [24], [25]. For wIRA with appropriate therapeutic irradiances and doses it could not only be demonstrated, that it is harmless for human skin (no induction of matrix metalloproteinase 1) [13], [24], but that it has cell protective effects against the damages caused by UV radiation [23], [24], [25]. (Safety aspects of the clinical use of wIRA are discussed as well in a separate section below.) In addition, wavelengths within wIRA have been shown to influence adhesive interactions between cells and extracellular matrices [9], playing a regulative role in wound repair processes, and may have a positive effect on cosmetic results [26]. It is also supposed that wIRA has immunomodulatory effects [1], [2].

Concerning both thermal and thermic as well as non-thermal and non-thermic effects of wIRA the mediation by pathways like nitric oxide in vasodilatation or by cytokines or neurotropines should also be taken into account [27].

Energy-related aspects of wound healing and oxygen

Wound healing and infection defense (e.g. function of granulocytes including their antibacterial oxygen radical formation) represent processes with an extremely high energy demand [1], [26], [27]. Hence they depend on a sufficient supply with energy and oxygen quite decisively. On the long run energy must be provided mostly aerobically (with oxygen). Oxygen plays a double role in wound healing: as an agent in the energy production and as a substrate for the oxygen radical formation of the granulocytes (respiratory burst) [1].

Wound repair and energy production therefore depend on the integrity of the following three vital factors [1], [26], [27]:

tissue temperature

tissue oxygen partial pressure

tissue perfusion.

Even one single factor lying clearly in the pathological area can deter energy production and wound healing or makes them both impossible [1], [26], [27]:

Below 28° C no wound healing is possible (too slow metabolism in accordance with the reaction velocity temperature rule) [1], [28] and the center of chronic wounds is often relatively hypothermic [1], [26], [27] – while e.g. both preoperative [1], [29] and postoperative [1], [26], [30] heat supply to the operation field can improve healing of acute wounds.

Without a sufficient oxygen partial pressure no aerobic energy production (and no granulocyte function) is possible (the center of chronic wounds frequently has an oxygen partial pressure near zero [1], [26], [27], [31], [32], [33], [34], [35], [36], [37], [38], [39], [40], [41]), which increases markedly the risk of wound infections [26].

A sufficient tissue blood flow including capillary blood flow is required for the transport of high-energy substrates to the tissue and for the removal of metabolic waste products [1].

The complex interaction of growth factors, cytokines, proteases and others in the context of wound healing with differences between acute and chronic wounds as well as influences of the age and the importance of the extracellular matrix – with e.g. an outweighing of inflammation mediators and proteases (matrix metalloproteinases) in chronic wounds – are today already quite well-known in detail [42], [43].

wIRA augments the cellular energy provision per time considerably by increasing all three factors, where the effects of wIRA on these three factors have been proven by different study groups by means of various methods [1], [26]:

Tissue temperature, proved in humans by means of a direct measuring of the tissue temperature with stitch probes [44], [45], [46] as well as with implanted probes (in 2 cm of tissue depth in operation wounds) [26] and thermographically [27], [47], [48] as well as in addition in animal experiments with stitch probes up to 7 cm of tissue depth [49];

e.g. wIRA can increase the temperature in 2 cm of tissue depth by approximately 2.7°C [26], the field of heat can reach into a depth of approximately 5(-7) cm [49].

Oxygen partial pressure in the tissue, proved in humans by means of a direct oxygen partial pressure measurement in the tissue with implanted probes in operation wounds [26] as well as by means of measuring of the oxygen saturation of the hemoglobin with an external white light-measuring probe [50];

e.g. wIRA can increase the oxygen partial pressure in 2 cm of tissue depth by approximately 30% [26].

Tissue blood flow/capillary blood flow, proved in humans by means of blood flow measurement with laser Doppler perfusion imaging (= scanning laser Doppler imaging) [47], [48], [51] and by means of blood flow measurement at two depths with an external laser Doppler-measuring probe [50] as well as in animal experiments by means of color microsphere technique up to 7 cm of tissue depth [49];

e.g. wIRA can increase superficial blood flow to approximately 8 times the amount [47], the field of increased blood flow can reach into a depth of approximately 5(-7) cm [49].

In contrast to this, hyperbaric oxygenation (HBO) [34], [35], [36] primarily increases only one factor, the oxygen partial pressure in the tissue.

The clinically beneficial effect of wIRA on problem wounds and wound infections – including the effects to decrease pain, inflammation, and hypersecretion and to have immunomodulatory effects – can be explained by the improvement in both the energy provision per time (increase of the metabolic rate) and the oxygen supply (e.g. for the function of granulocytes) as well as by non-thermal and non-thermic cellular effects [1], [13].

Due to its penetration properties, wIRA allows a multiple energy transfer into subcutaneous tissue (2-3 cm) without irritating or overheating the skin like unfiltered heat radiation [1], [2], [26]. As many postoperative wound healing impairments and infections originate primarily in the subcutaneous layer, wIRA has advantages for local warming in acute wound healing compared to other sources such as heating bandage systems or hot packs, whose heat is absorbed in the epidermal layers and may cause burning of the skin [26], [52].

Taking the holistic point of view of quantum physics [53] into account, water-filtered infrared-A can be described as flow of photons (quanta) both with non-thermal and non-thermic as well as with thermal and thermic effects [13]. From the point of view of modern physics with its probabilistic approach [53], [54], regarding the interaction of elements within a system (with an irreversibility of time and a sequence of events and small influences leading to divergent ways and results (butterfly phenomenon)), many systems in the world, especially biological systems, are unstable thermodynamic systems, capable to build up and represent complex structures and being far away from a stable (unstructured) point (chaos) [13]. Energy delivery to the system can maintain such an unstable thermodynamic system [13], [54]. In this sense an adequate infrared irradiation with appropriate irradiances can help maintain such a desired unstable thermodynamic system: on the macroscopic level predominantly with thermal (transfer of energy) and thermic effects (clinically with increased tissue temperature, perfusion and tissue oxygen partial pressure as energy-related important variables [26]) and on the microscopic/molecular level both with non-thermal and non-thermic as well as with thermal and thermic effects on cells and cell structures [13].

Principles of clinical applications of wIRA

wIRA can always be taken into consideration when a depth effective heat application is desired/indicated clinically (with good tolerance to high power density and with a high energy flow into the tissue). wIRA can always be taken into account when pathogenetically at least one factor which can be influenced positively by the thermal and thermic as well as by non-thermal and non-thermic effects of wIRA is impaired or suboptimal [1].

Advantages of wIRA

Decrease of pain, inflammation, and hypersecretion and positive immunomodulatory effects [1]. All four effects are clinically important. Especially the pain reduction (or the pruritus reduction in morphea [55], [56]), seen in a variety of indications, e.g. in verrucae [2], herpes, wounds [1], [26], [27], [57], [58], [59], scleroderma [60], and observed in different study groups, with its positive consequences for the patients (less pain, remarkably less need for analgesics, less side-effects of analgesics) should be emphasized as an important clinical effect of wIRA.

contact-free, easily used procedure

"clean" procedure (compared to e.g. fango)

without expenditure of material

usable for a single body region (single radiator)

gentle concerning blood circulation (compared with full bath)

no need for a fixing at the body (compared with a "warm pack")

usable at all sorts of positionings

offers freedom of movement

possible combination of "heat and motion" [61]

ideal dosing properties (dosing primarily by variation of the distance from the radiator)

continuously rising temperature without heat shock and overheating of the superficial skin layers

subjectively pleasant (even on wounds), therefore unproblematic use also with children

good effects in the depth

long lasting heat depot

relatively low technical expenditure

low time expenditure for staff

easy feasibility

limited time expenditure for the patient

Altogether, wIRA is:

fundamentally better than "red light" (unfiltered infrared), because a considerably higher irradiance is possible with more warming in the depth and less heating of the surface and

also a better alternative to "wet warm packs" and other heating methods.

Beside the possibility that wIRA radiators are used in hospitals or in offices of physicians or surgeons, wIRA radiators can be used – under the supervision of the responsible physician – directly at home or similarly in a nursing home: especially when for a longer period of time a wound shall be irradiated once or twice daily and the patient itself or his family or an ambulatory nursing care service takes care to use the radiator appropriately, a wIRA radiator can be provided on loan [1].

Fundamental recommendations for the clinical use of wIRA

The following fundamental recommendations can be given for clinical use of wIRA:

Typically wIRA acts only on bare skin, i.e. wIRA does not penetrate clothes or most kinds of bandages or wound dressings.

If possible, irradiation should be vertical to the skin; irradiation distance should be at least as indicated by the distance rod of the radiator (depends on the type of radiator, e.g. 25 cm) [1].

Irradiation time should be at least 20 minutes (better 30 minutes) [1].

If the patient gets too warm, at full term the irradiation distance should be extended (and by this the irradiance decreased); if possible, the irradiation should not be stopped [1].

In routine clinical practice often markedly larger irradiation distances than the minimum distance are used. Then the typical total irradiances (wIRA and visible light) are approximately 80-160 mW/cm² (depending on size of the irradiated area, on tissue temperature and amount of subcutaneous soft tissues, e.g. lower used irradiances at the tibial border compared to the anterior part of the thigh), corresponding to wIRA irradiances of only 60-120 mW/cm² [13].

Special caution, i.e. a larger irradiation distance, should be taken in patients with an impaired sensation (e.g. diabetic polyneuropathy) or a deteriorated ability to express themselves, and when irradiating cold tissue or tissue badly supplied with blood or an area with low subcutaneous tissue (e.g. tibial border) [1].

wIRA for the improvement of wound healing in dermatology and surgery

Acute wounds and especially chronic wounds, intractable wounds or infected problem wounds should be irradiated with wIRA ideally once or twice per day for (20-)30 minutes each (longer irradiation times per day are possible and often helpful), at least three times per week for (20-)30 minutes [1]. wIRA does not replace other sensible/necessary therapeutic procedures (such as the important compression garment therapy of chronic venous stasis ulcers of the lower legs [62], [63], [64]) but complements them. Correspondingly the therapy with wIRA has to be embedded in an overall therapeutic concept. wIRA can be used independently from therapy preferences concerning wound management (e.g. moist wound management). Typically for wIRA irradiation the wound has to be uncovered, see Figure 6 (Fig. 6), as most bandages or wound dressings (with the exception of e.g. some tested transparent foils) are not adequately permeable for wIRA (as demonstrated by spectral transmission measurements) [1].

According to modern concepts [65] for the assessment of wound healing also other end-points and variables of interest aside from a complete wound closure have to be used like reduction of pain, improvement of quality of life, improvement of the cosmetic result, reduction of scars, clinically relevant shortening of the time of wound healing and improved quality of healing [1]. Nowadays great importance is placed on the reduction or avoidance of pain in order to improve the wound healing and to avoid the formation of a pain memory with chronification of the pain [66], [67] associated with the application of management strategies of common acute and chronic wounds.

Safety aspects of the clinical use of wIRA

wIRA in clinical use at appropriate irradiances has been described since more than 15 years as helpful and safe [1], [2], [23], [24], [26], [27], [34], [44], [45], [46], [47], [48], [49], [55], [57], [58], [59], [61], [68], [69], [70], [71], [72], [73], [74], [75], [76], [77], [78], [79], [80], [81].

In accordance with previous investigations [19], [20], [21] water-filtered infrared-A (wIRA) at appropriate irradiances is unlike ultraviolet-A especially not implicated in photoaging of the skin, mediated by the collagenase matrix metalloproteinase 1 (MMP-1) [13], [23], [24]. wIRA can even be implicated in a protective manner [13], [19], [20], [22], [23], [24].

Investigations of human skin fibroblasts after single exposures between 15 minutes and 8 hours to wIRA or 15-45 minutes to ultraviolet-A (UV-A) radiation at two physiologic temperatures as well as after repeated exposures with wIRA are presented in [13]: Single exposure of cultured human dermal fibroblasts to UV-A radiation yielded a very high increase in MMP-1 mRNA expression (11-fold expression for conventional Reverse Transcriptase Polymerase Chain Reaction (RT-PCR) and 76-fold expression for quantitative real-time RT-PCR) and a dose-dependent decrease in cell survival. In contrast, even at an investigated disproportionally high irradiance wIRA did not produce cell death and did not induce an increase in MMP-1 mRNA expression. Additionally, repeated exposure to wIRA did not induce MMP-1 mRNA expression.

The results of the recent publication [13], in which particularly a great effort to avoid any temperature alteration of the cells was undertaken, are consistent with publications of several study groups [13], [19], [20], [21], [22], [23], [24], [82] and contradict publications primarily of one study group [83], [84], [85], [86], which described undesired effects of infrared-A and wIRA radiation in cell cultures.

Reasons for the different results are presented in detail in [13]: so [13] used normal skin fibroblasts and not foreskin fibroblasts of the newborn or young child which are well known to behave differently and are not representative of human skin. Effects on cells depend not only on the irradiation dose but also on the irradiance. In some other publications it was not taken into account that cell cultures, which represent a monolayer of cells and which have in contrast to human skin no horny layer and no heat distribution by blood circulation, cannot be irradiated with the same irradiance as a living skin (the remarkable difference between irradiating a material without circulation and a living patient is described and documented, for example, in two infrared thermography video sequences in [27]): corresponding to this partly the three- to tenfold (e.g. 333 mW/cm² [83]) of a physiologic irradiance was used in cell cultures [83], [84], [85], [86]. In [13] with careful and effortful temperature fixing of the cells, which prevented an overheating of the cells, no undesired thermic effects were seen even at the investigated disproportionally high irradiance. Such a temperature fixing is important to avoid misinterpretations, as MMP-1 mRNA, whose increase is interpreted partly as an indication for skin aging, shows an increased expression already by temperature increase about a threshold – a pure thermic effect independent from the cause of the temperature increase and thus also independent from specific properties of infrared-A or wIRA radiation. Interestingly a current publication with emphasis to avoid any overheating [82] from the same institute as the publications [83], [84], [85], [86] showed no damaging effect of water-filtered infrared-A radiation (0, 100, 250, 500, 1000 J/cm²) to a glyceraldehyde-3-phosphate dehydrogenase (GAPDH) solution as a model enzyme for environmentally induced protein damage, while irradiation with UV-A (0, 100, 250, 500, 1000 J/cm²), UV-B (0, 250, 500, 1000, 2000 mJ/cm²), and gamma-irradiation (0, 50, 100, 250, 500 Gy) caused a dose-dependent increase in protein modification (fragmentation and aggregation) and loss of enzyme activity with complete loss of enzyme activity at the highest doses. The striking difference between appropriate and inappropriate irradiances can be illustrated by the maximum skin temperature: Mercer [47] showed that a clinically typical irradiation with wIRA increased skin surface temperature (starting from 32.5°C) by nearly 6°C to a mean of 38.2°C (maximum value was 39.1°C). In accordance with that skin surface temperature (starting with approximately 32°C) reaches approximately only 38°C even under Mediterranean conditions in the summer at noon after 30 minutes stay in the sun [13]. These approximately 38(-39)°C are decisively lower than the induced “43°C for 90 minutes” with inappropriately high irradiance [87]: above 39.5-40°C heat-shock proteins can be induced [13] (the heat-shock-induced matrix metalloproteinase-1 expression in human epidermal keratinocytes is mediated by the transient receptor potential vanilloid-1 kation channel [88]), and this is a thermal effect and especially a thermic effect (temperature limits are exceeded), independent from the cause of exceeding temperature limits, and not a direct radiation effect. Some more examples of inappropriate irradiances in vivo and in vitro are presented in [13].

Appropriate therapeutic irradiances of wIRA – typically within the range 60-120 mW/cm² – are clearly lower than the above described investigated disproportionally high irradiance (333 mW/cm² [83]), particularly if the difference between living skin and cell culture is taken into account. The infrared-A irradiance by the sun at the surface of the earth on sea-level during the summer at noontime in moderate climatic zones is once more considerably lower (maximum approximately 20 mW/cm²) – and even under extreme atmospheric conditions at the equator only approximately 34 mW/cm². Therefore related to infrared-A irradiances and doses no indications for a necessary or sensible protection of the skin against infrared-A or wIRA radiation of an appropriate therapeutic use or of the sun can be seen [13] in contrast to the undisputedly sensible and necessary protection against an excessive UV irradiation.

As explained in the section about thermal and thermic effects of wIRA, wIRA with its special properties allows a high energy transfer into tissue (with relevant desired thermal effects, like increase of tissue temperature, tissue oxygen partial pressure, and tissue perfusion) combined with a limited temperature increase in the tissue (with limited thermic effects, omitting undesired effects) and with ideal dosing properties.

Clinical indications for wIRA

A review concerning clinical indications for wIRA in acute wounds including prospective, randomized, controlled, double-blind studies is presented in [89].

A review concerning clinical indications for wIRA in chronic wounds including prospective, randomized, controlled, double-blind studies is presented in [14].

Clinical indications for wIRA outside wound healing are [1]:

In dermatology: wIRA alone (which means without simultaneously topically administered substance and without the subject of photodynamic therapy PDT) can be used for therapeutic purposes in recalcitrant common hand and foot warts (one therapy cycle with continous keratolysis with salicylic acid plaster, bloodless curettage, one wIRA irradiation of 30 minutes per week for 6-9 weeks [2], [55]), in herpes labialis, herpes zoster (fast decrease of pain in the acute phase as observed in casuistics), condylomata acuminata, scleroderma [60], [72], morphea [55], [56], acne papulopustulosa [90], [91], and possibly in alopezia areata.

wIRA can be used to improve the penetration of topically applied substances [69], [70], [71] (like cortisone or local anaesthetics) as an alternative to an occlusive dressing. This can be considered (indications under investigation) in neurodermitis, psoriasis, herpes zoster (with acyclovir topically), acne papulopustulosa (with topical acne therapeutic [91]), alopezia areata (with topical cortisone).

wIRA can be used within a photodynamic therapy PDT (and a photodynamic diagnosis PDD = fluorescence diagnosis FD) – if clinically an indication for a PDT or a PDD with PDT is given – together with one or several absorption bands in the visible range in actinic keratosis [73], [74] and in (superficial) basal cell carcinomas (BBC) [74], [92]. Perspectives for a use of PDT (with wIRA) in wound healing are presented in [14].

In physiotherapy, sports medicine, and orthopedics: the clinical application of wIRA can be in preventive, therapeutic, regenerative, or rehabilitative intention. Muscular hardenings, myogeloses [66], lumbago, diseases of the rheumatic disorders circle [76], M. Bechterew [77], arthroses, arthritises, contusions; fibromyalgia (preferably wIRA in combination with motion, i.e. wIRA with a small amount of ergometer work) [93], regeneration after sports [78] (wIRA alone or wIRA in combination with motion), postoperative rehabilitation [89], [94], improvement of lipolysis (wIRA in combination with motion) [61].

In neonatology: keeping or rising of the body temperature, generating a "heat depot" ahead of a necessary transport of a neonate [79].

In oncology: (local or systemic) hyperthermia in combination with radiotherapy (e.g. in metastatic breast cancer [95]), hyperthermia in combination with chemotherapy [96].

Figure 1View larger versionFigure 1. Spectral solar irradiance outside the atmosphere and at sea level in both cases with the sun at the zenith and for a mean Earth-sun separation. Shaded areas indicate absorption at sea level due to the atmospheric constituents shown (from [97], adapted from [98]).

For comparison of Figures 1, 3 and 4: 1000 W m⁻² µm⁻¹ = 100 mW cm⁻² µm⁻¹ = 1 mW cm⁻² (10 nm)⁻¹

Figure 2View larger versionFigure 2. Cross-section of a water-filtered infrared-A radiator (Hydrosun, Müllheim, Germany)

The whole incoherent broad-band radiation of a 3000 Kelvin halogen bulb is passed through a cuvette, containing water, which absorbs or decreases the undesired wavelengths within infrared (most parts of infrared-B and -C and the absorption bands of water within infrared-A). The water is hermetically sealed within the cuvette. A fan provides air cooling of the cuvette to prevent the water from boiling.

Figure 3View larger versionFigure 3. Spectral irradiance of a water-filtered infrared-A radiator (Hydrosun® radiator 501 with 10 mm water cuvette and orange filter OG590) at approximately 210 mW/cm² (= 2.1 x 10³ W/m²) total irradiance (from [1]); (visible light (VIS): 380-780 nm; infrared-A (IR-A): 780-1400 nm; infrared-B (IR-B): 1400-3000 nm)

Figure 4View larger versionFigure 4. Comparison of the spectra of the sun at sea level and of a water-filtered infrared-A radiator Spectral solar irradiance at sea level (with the sun at the zenith and for a mean Earth-sun separation) as in Fig. 1 (adapted from [97]) and spectral irradiance of a water-filtered infrared-A radiator (Hydrosun® radiator 501 with 10 mm water cuvette and orange filter OG590) at approximately 210 mW/cm² (= 2.1 x 10³ W/m²) total irradiance as in Fig. 3 (from [1]).

The spectrum of the sun at sea level includes ultraviolet radiation (UV, <400 nm), visible light (VIS, 380-780 nm), and infrared radiation (IR, >780 nm). The spectrum of the water-filtered infrared-A radiator includes only visible light (VIS) and infrared radiation (IR); the visible part depends on the used color filter; the wIRA radiator does not emit ultraviolet radiation (UV).

Both spectra show the decreased irradiance of the absorption bands of water.

Figure 5View larger versionFigure 5. Comparison of irradiation with water-filtered infrared-A and with conventional infrared Thermographical comparison of skin surface temperatures in the lumbar region 12 minutes after beginning of irradiation with water-filtered infrared-A (left) and conventional infrared (right) with the same irradiance: the skin surface temperature is higher in case of irradiation with conventional infrared (presented in the thermography), while temperature in 1 cm depth of tissue is higher when irradiating with water-filtered infrared-A (from [45]). So water-filtered infrared-A presents a high tissue penetration combined with a low thermal load to the skin surface.

Figure 6View larger versionFigure 6. Example for an irradiation of a wound with a water-filtered infrared-A radiator (published with kind approval of Prof. James Mercer, Tromsø/Norway) [27]

Articles from GMS Krankenhaushygiene interdisziplinär are provided here courtesy of German Medical Science

PMC Copyright Notice

The articles available from the PMC site are protected by copyright, even though access is free. Copyright is held by the respective authors or publishers who provide these articles to PMC. Users of PMC are responsible for complying with the terms and conditions defined by the copyright holder.

Users should assume that standard copyright protection applies to articles in PMC, unless an article contains an explicit license statement that gives a user additional reuse or redistribution rights. PMC does not allow automated/bulk downloading of articles that have standard copyright protection.

See the copyright notice on the PMC site, https://www.ncbi.nlm.nih.gov/pmc/about/copyright/, for further details and specific exceptions.

References

Hoffmann G. Wassergefiltertes Infrarot A (wIRA) zur Verbesserung der Wundheilung [Übersichtsarbeit] [Water-filtered infrared A (wIRA) for the improvement of wound healing [review]]. GMS Krankenhaushyg Interdiszip. 2006;1(1):Doc20 . Available from: http://www.egms.de/en/journals/dgkh/2006-1/dgkh000020.shtml.

Fuchs SM, Fluhr JW, Bankova L, Tittelbach J, Hoffmann G, Elsner P. Photodynamic therapy (PDT) and waterfiltered infrared A (wIRA) in patients with recalcitrant common hand and foot warts. Ger Med Sci. 2004;2:Doc08 . Available from: http://www.egms.de/en/gms/2004-2/000018.shtml. [PubMed]

Cobarg CC, author; Vaupel P, Krüger W, editors. Physikalische Grundlagen der wassergefilterten Infrarot-A-Strahlung [Principles of the physical properties of water-filtered infrared-A radiation]. Wärmetherapie mit wassergefilterter Infrarot-A-Strahlung [Thermal therapy with water-filtered infrared-A radiation]. 1995. 2. Aufl. Stuttgart: Hippokrates; p. 19–28

Rzeznik J, author; Vaupel P, Krüger W, editors. Die Technik zur loko-regionalen Wärmetherapie mit wassergefilterter Infrarot-A-Strahlung [The technique of loco-regional thermal therapy with water-filtered infrared-A radiation]. Wärmetherapie mit wassergefilterter Infrarot-A-Strahlung [Thermal therapy with water-filtered infrared-A radiation]. 1995. 2. Aufl. Stuttgart: Hippokrates; p. 29–46

Albrecht-Buehler G. Surface extensions of 3T3 cells towards distant infrared light sources. J Cell Biol. 1991;114(3):493–502. [PubMed]

Albrecht-Buehler G. Cellular infrared detector appears to be contained in the centrosome. Cell Motil Cytoskeleton. 1994;27(3):262–271. [PubMed]

Albrecht-Buehler G. A long-range attraction between aggregating 3T3 cells mediated by near-infrared light scattering. Proc Natl Acad Sci U S A. 2005;102(14):5050–5055. [PubMed]

Ehrlicher A, Betz T, Stuhrmann B, Koch D, Milner V, Raizen MG, Käs J. Guiding neuronal growth with light. Proc Natl Acad Sci USA. 2002;99(25):16024–16028. . Available from: http://dx.doi.org/10.1073/pnas.252631899. [PubMed]

Karu TI, Pyatibrat LV, Kalendo GS. Cell attachment to extracellular matrices is modulated by pulsed radiation at 820 nm and chemicals that modify the activity of enzymes in the plasma membrane. Lasers Surg Med. 2001;29(3):274–281. [PubMed]

Karu TI, Pyatibrat LV, Kalendo GS. Donors of NO and pulsed radiation at lambda = 820 nm exert effects on cell attachment to extracellular matrices. Toxicol Lett. 2001;121(1):57–61. [PubMed]

Karu TI, Pyatibrat LV, Kalendo GS. Cell attachment modulation by radiation from a pulsed light diode (lambda = 820 nm) and various chemicals. Lasers Surg Med. 2001;28(3):227–236. [PubMed]

Chow RT, Heller GZ, Barnsley L. The effect of 300 mW, 830 nm laser on chronic neck pain: a double-blind, randomized, placebo-controlled study. Pain. 2006;124(1-2):201–210. [PubMed]

Gebbers N, Hirt-Burri N, Scaletta C, Hoffmann G, Applegate LA. Water-filtered infrared-A radiation (wIRA) is not implicated in cellular degeneration of human skin. GMS Ger Med Sci. 2007;5:Doc08 . Available from: http://www.egms.de/en/gms/2007-5/000044.shtml.

von Felbert V, Schumann H, Mercer JB, Strasser W, Daeschlein G, Hoffmann G. Therapy of chronic wounds with water-filtered infrared-A (wIRA) [review]. GMS Krankenhaushyg Interdiszip. 2007;2(2):Doc52 . Available from: http://www.egms.de/en/journals/dgkh/2008-2/dgkh000085.shtml.

Karu TI. Primary and secondary mechanisms of action of visible to near-IR radiation on cells [review]. J Photochem Photobiol B. 1999;49(1):1–17. [PubMed]

Karu TI, author; Waynant RW, editor. Low-power laser effects. Lasers in medicine. 2002. Boca Raton: CRC Press; p. 171–209

Danno K, Mori N, Toda K, Kobayashi T, Utani A. Near-infrared irradiation stimulates cutaneous wound repair: laboratory experiments on possible mechanisms. Photodermatol Photoimmunol Photomed. 2001;17(6):261–265. [PubMed]

Horwitz LR, Burke TJ, Carnegie D. Augmentation of wound healing using monochromatic infrared energy. Exploration of a new technology for wound management. Adv Wound Care. 1999;12(1):35–40. [PubMed]

Menezes S, Coulomb B, Lebreton C, Dubertret L. Non-coherent near infrared radiation protects normal human dermal fibroblasts from solar ultraviolet toxicity. J Invest Dermatol. 1998;111(4):629–633. [PubMed]

Frank S, Menezes S, Lebreton-De Coster C, Oster M, Dubertret L, Coulomb B. Infrared radiation induces the p53 signaling pathway: role in infrared prevention of ultraviolet B toxicity. Exp Dermatol. 2006;15(2):130–137. [PubMed]

Frank S, Oliver L, Lebreton-De Coster C, Moreau C, Lecabellec MT, Michel L, Vallette FM, Dubertret L, Coulomb B. Infrared radiation affects the mitochondrial pathway of apoptosis in human fibroblasts. J Invest Dermatol. 2004;123(5):823–831. [PubMed]

Danno K, Horio T, Imamura S. Infrared radiation suppresses ultraviolet B-induced sunburn-cell formation. Arch Dermatol Res. 1992;284(2):92–94. [PubMed]

Applegate LA, Scaletta C, Panizzon R, Frenk E, Hohlfeld P, Schwarzkopf S. Induction of the putative protective protein ferritin by infrared radiation: implications in skin repair. Int J Mol Med. 2000;5(3):247–251. [PubMed]

Burri N, Gebbers N, Applegate LA, authors; Pandalai SG, editor. Chronic infrared-A radiation repair: Implications in cellular senescence and extracellular matrix. Recent Research Developments in Photochemistry & Photobiology. 2004. 7. Trivandrum: Transworld Research Network; p. 219–231

Hoffmann G, Meffert H. Apparent contradiction between negative effects of UV radiation and positive effects of sun exposure. GMS Ger Med Sci. 2005;3:Doc01 . Available from: http://www.egms.de/en/gms/2005-3/000019.shtml.

Hartel M, Hoffmann G, Wente MN, Martignoni ME, Büchler MW, Friess H. Randomized clinical trial of the influence of local water-filtered infrared A irradiation on wound healing after abdominal surgery. Br J Surg. 2006;93(8):952–960. [Cross Ref] . Available from: http://dx.doi.org/10.1002/bjs.5429. [PubMed]

Mercer JB, Nielsen SP, Hoffmann G. Improvement of wound healing by water-filtered infrared-A (wIRA) in patients with chronic venous leg ulcers including evaluation using infrared thermography. GMS Ger Med Sci. 2008;6:

Hartel W, author. Personal communication. 2001

Melling AC, Ali B, Scott EM, Leaper DJ. Effects of preoperative warming on the incidence of wound infection after clean surgery: a randomised controlled trial. Lancet. 2001;358:876–880. [PubMed]

Plattner O, Akca O, Herbst F, Arkilic CF, Függer R, Barlan M, Kurz A, Hopf H, Werba A, Sessler DI. The influence of 2 surgical bandage systems on wound tissue oxygen tension. Arch Surg. 2000;135:818–822. [PubMed]

Kivisaari J, Vihersaari T, Renvall S, Niinikoski J. Energy metabolism of experimental wounds at various oxygen environments. Ann Surg. 1975;181:823–828. [PubMed]

Kühne HH, Ullmann U, Kühne FW. New aspects on the pathophysiology of wound infection and wound healing - the problem of lowered oxygen pressure in the tissue. Infection. 1985;13:52–56. [PubMed]

Niinikoski J, Gottrup F, Hunt TK, authors; Janssen H, Rooman R, Robertson JIS, editors. The role of oxygen in wound repair. Wound healing. 1991. Petersfield: Wrightson Biomedical Publishing;

Hoffmann G. Improvement of wound healing in chronic ulcers by hyperbaric oxygenation and by waterfiltered ultrared A induced localized hyperthermia. Adv Exp Med Biol. 1994;345:181–188. [PubMed]

Buslau M, Hoffmann G, authors; Fuchs J, Packer L, editors. Hyperbaric oxygenation in the treatment of skin diseases [review]. Oxidative stress in dermatology. 1993. New York: Marcel Dekker; p. 457–485

Buslau M, Hoffmann G. Die hyperbare Oxygenation (HBO) - eine adjuvante Therapie akuter und chronischer Wundheilungsstörungen [Hyperbaric oxygenation - an adjuvant therapy of acute and chronic wound healing impairments]. Dermatol Monatsschr. 1993;179:39–54

Hoffmann G, Buslau M, authors; Cramer FS, editor. Treatment of skin diseases by hyperbaric oxygenation. Proceedings of the Eleventh International Congress on Hyperbaric Medicine. 1995. Flaggstaff, USA: Best Publishing Company; p. 20–21. p. 153–159

Wright J, author. Hyperbaric oxygen therapy for wound healing. World Wide Wounds. 2001. . Available from: http://www.worldwidewounds.com/2001/april/Wright/HyperbaricOxygen.html.

Knighton DR, Silver IA, Hunt TK. Regulation of wound-healing angiogenesis - effect of oxygen gradients and inspired oxygen concentration. Surgery. 1981;90:262–270. [PubMed]

Barnikol WKR, Teslenko A, Pötzschke H. Eine neue topische Behandlung chronischer Wunden mit Haemoglobin und Sauerstoff: Verfahren und erste Ergebnisse [A new topic treatment of chronic wounds with haemoglobin and oxygen: procedere and first results]. Z Wundheilung - J Wound Healing. 2005;10(3):98–108

Jünger M, Hahn M, Klyscz T, Steins A, authors. Role of microangiopathy in the development of venous leg ulcers. 1999. 23. Basel: Karger; p. 180–193. (Progr. Appl. Microc.)

Enoch P, Price P, authors. Cellular, molecular and biochemical differences in the pathophysiology of healing between acute wounds, chronic wounds and wounds in the aged. World Wide Wounds. 2004. . Available from: http://www.worldwidewounds.com/2004/august/Enoch/Pathophysiology-Of-Healing.html.

Schultz GS, Ludwig G, Wysocki A, authors. Extracellular matrix: review of its roles in acute and chronic wounds. World Wide Wounds. 2005. . Available from: http://www.worldwidewounds.com/2005/august/Schultz/Extrace-Matric-Acute-Chronic-Wounds.html.

Vaupel P, Stofft E, authors; Vaupel P, Krüger W, editors. Wassergefilterte Infrarot-A-Strahlung im Vergleich zu konventioneller Infrarotstrahlung oder Fango-Paraffin-Packungen: Temperaturprofile bei lokaler Wärmetherapie [Water-filtered infrared-A radiation in comparison to conventional infrared-A radiation or fango paraffine packages: temperature profiles in local thermal therapy]. Wärmetherapie mit wassergefilterter Infrarot-A-Strahlung [Thermal therapy with water-filtered infrared-A radiation]. 1995. 2. Aufl. Stuttgart: Hippokrates; p. 135–147

Vaupel P, Rzeznik J, Stofft E. Wassergefilterte Infrarot-A-Strahlung versus konventionelle Infrarotstrahlung: Temperaturprofile bei lokoregionaler Wärmetherapie [Water-filtered infrared-A radiation versus conventional infrared-A radiation: temperature profiles in local thermal therapy]. Phys Med Rehabilitationsmed Kurortmed. 1995;5:77–81

Stofft E, Vaupel P. Wassergefilterte Infrarot-A-Strahlung versus Fango-Paraffin-Packung: Temperaturprofile bei lokoregionaler Wärmetherapie [Water-filtered infrared-A radiation versus fango paraffine packages: temperature profiles in local thermal therapy]. Phys Med Rehabilitationsmed Kurortmed. 1996;6:7–11

Mercer JB, de Weerd L. The effect of water-filtered infrared-A (wIRA) irradiation on skin temperature and skin blood flow as evaluated by infrared thermography and scanning laser Doppler imaging. Thermology Int. 2005;15(3):89–94

Pascoe DD, Mercer JB, de Weerd L, authors. Physiology of thermal signals. Biomedical Engineering Handbook. 2006. 3rd edition. Boca Raton (Florida/USA): Taylor and Francis Group, CRC press; p. 21–1

Hellige G, Becker G, Hahn G, authors; Vaupel P, Krüger W, editors. Temperaturverteilung und Eindringtiefe wassergefilterter Infrarot-A-Strahlung [Temperature distribution and penetration depth of water-filtered infrared-A radiation]. Wärmetherapie mit wassergefilterter Infrarot-A-Strahlung [Thermal therapy with water-filtered infrared-A radiation]. 1995. 2. Aufl. Stuttgart: Hippokrates; p. 63–79

Schumann H, Schempp CM, authors. wIRA in der Wundtherapie - Erste Erfahrungen in der Anwendung bei chronischen Wunden in der Universitäts-Hautklinik Freiburg, Lecture presented at the symposium entitled "Water-filtered infrared-A (wIRA) in dermatology" of the Dr. med. h.c. Erwin Braun Foundation in Liestal/Basel, Switzerland, November 20, 2004. [wIRA in wound therapy - first experiences in the application in chronic wounds in the Department of Dermatology of the University Hospital Freiburg]. 2004

Heinemann C, Elsner P, authors. Personal communication. 2002

Toyokawa H, Matsui Y, Uhara J, Tsuchiya H, Teshima S, Nakanishi H, et al.Promotive effects of far-infrared ray on full-thickness skin wound healing in rats. Exp Biol Med (Maywood). 2003;228:724–729. [PubMed]

von Weizsäcker CF, author. Die Philosophie eines Physikers. Lecture in the TV series Tele-Academy. 3sat [Germany, Austria, Switzerland]. 08.04.2007, 07.15-08.00. [Lecture primarily presented 1992 at the University Bamberg/Germany.] [The philosophy of a physicist]. 2007

Prigogine I, author. Chaos and the Law of Nature. Lecture in the TV series Tele-Academy. 3sat [Germany, Austria, Switzerland]. 29.04.2007, 07.15-08.00. [Lecture of the president of the International Academy of Science primarily presented 1994 in Stuttgart/Germany.] [Chaos und Naturgesetze]. 2007

von Felbert V, Streit M, Weis J, Braathen LR. Anwendungsbeobachtungen mit wassergefilterter Infrarot-A-Strahlung in der Dermatologie [Application observations with water-filtered infrared-A radiation in dermatology]. Dermatol Helvetica. 2004;16(7):32–33

von Felbert V, Hunziker T, Simon D, Braathen LR, authors. Water-filtered infrared-A irradiation for cutaneous scleroderma. A report of four cases. [Publication in preparation].

Biland L, Barras J. Die wassergefilterte Infrarot-A-Hyperthermie zur Behandlung venöser Ulcera [Water-filtered infrared-A induced hyperthermia used as therapy of venous ulcers]. Hefte Wundbehand. 2001;5:41

Hoffmann G. Water-filtered infrared A (wIRA) for the improvement of wound healing in acute and chronic wounds. Z Wundheilung - J Wound Healing. 2005;(special issue 2):130

Hoffmann G. Wassergefiltertes Infrarot A (wIRA) zur Verbesserung der Wundheilung bei akuten und chronischen Wunden [Water-filtered infrared-A (wIRA) for the improvement of wound healing of acute and chronic wounds]. MedReport. 2005;29(34):4 . Available from: http://www.medreports.de/medpdf05/mreport34_05.pdf.

Buslau M, author. Wassergefiltertes Infrarot A (wIRA) bei Sklerodermie. (Ergebnisse einer prospektiven, randomisierten, kontrollierten, doppelt-blinden klinischen Studie.) [Water-filtered infrared-A (wIRA) in scleroderma. (Results of a prospective, randomized, controlled, double-blind study.)] Lecture presented at the course entitled "Water-filtered infrared-A in dermatology" of the Department of Dermatology of the University Hospital Jena in Jena, Germany, November 15, 2003. 2003

Möckel F, Hoffmann G, Obermüller R, Drobnik W, Schmitz G. Influence of water-filtered infrared-A (wIRA) on reduction of local fat and body weight by physical exercise. GMS Ger Med Sci. 2006;4:Doc05 . Available from: http://www.egms.de/en/gms/2006-4/000034.shtml.

Thomas S, author. Compression bandaging in the treatment of venous leg ulcers. World Wide Wounds. 1997 First publication 1997, last modification 2001. Available from: http://www.worldwidewounds.com/1997/september/Thomas-Bandaging/bandage-paper.html.

Johnson S, author. Compression hosiery in the prevention and treatment of venous leg ulcers. World Wide Wounds. 2002. . Available from: http://www.worldwidewounds.com/2002/september/Johnson/Compression-Hosiery-Leg-Ulcers.html.

Thomas S, Fram P, Phillips P, authors. The importance of compression on dressing performance. World Wide Wounds. 2007. . Available from: http://www.worldwidewounds.com/2007/November/Thomas-Fram-Phillips/Thomas-Fram-Phillips-Compression-WRAP.html.

Enoch P, Price P, authors. Should alternative endpoints be considered to evaluate outcomes in chronic recalcitrant wounds? World Wide Wounds. 2004. . Available from: http://www.worldwidewounds.com/2004/october/Enoch-Part2/Alternative-Enpoints-To-Healing.html.

Hoffmann G, Siegfried I, authors. Volkskrankheit Rückenschmerz: neue Sichtweisen. Seminar des Arbeitskreises Sportmedizin der Akademie für ärztliche Fortbildung und Weiterbildung der Landesärztekammer Hessen. Bad Nauheim, 05.06.2004 [Common illness backache: new ways of looking at]. 2004. Düsseldorf, Köln: German Medical Science; p. Doc 04ruecken1 . Available from: http://www.egms.de/en/meetings/ruecken2004/04ruecken1.shtml.

Pediani R, author. What has pain relief to do with acute surgical wound healing? World Wide Wounds. 2001. . Available from: http://www.worldwidewounds.com/2001/march/Pediani/Pain-relief-surgical-wounds.html.

Illing P, Gresing T. Improvement of wound healing in severely burned children by water-filtered infrared-A (wIRA). GMS Ger Med Sci. 2008;6:

Haupenthal H, author. In vitro- und in vivo-Untersuchungen zur temperaturgesteuerten Arzneistoff-Liberation und Permeation [Thesis] [In vitro and in vivo investigations of temperature dependent drug liberation and permeation]. 1997. Mainz: Johannes Gutenberg-Universität;

Bankova L, Heinemann C, Fluhr JW, Hoffmann G, Elsner P, authors. Improvement of penetration of a topical corticoid by waterfiltered infrared A (wIRA). In: 2003. 1st Joint Meeting 14th International Congress for Bioengineering and the Skin & 8th Congress of the International Society for Skin Imaging; 2003 May 21-24; Hamburg. p. P96

Otberg N, Grone D, Meyer L, Schanzer S, Hoffmann G, Ackermann H, Sterry W, Lademann J. Water-filtered infrared-A (wIRA) can act as a penetration enhancer for topically applied substances. GMS Ger Med Sci. 2008;6:

Meffert H, Buchholtz I, Brenke A. Milde Infrarot-A-Hyperthermie zur Behandlung der systemischen Sklerodermie [Mild infrared-A hyperthermia for treatment of systemic scleroderma]. Dermatol Monatsschr. 1990;176(11):683–686. [PubMed]

Foss P. Einsatz eines patentierten, wassergefilterten Infrarot-A-Strahlers (Hydrosun) zur photodynamischen Therapie aktinischer Dyskeratosen der Gesichts- und Kopfhaut [Application of a patented water-filtered infrared-A radiator (Hydrosun) for photodynamic therapy of actinic keratosis of the skin of the face and the scalp]. Z naturheilkundl Onkologie krit Komplementärmed. 2003;6(11):26–28

Hübner K. Die Photo-dynamische Therapie (PDT) der aktinischen Keratosen, Basalzellkarzinome und Plantarwarzen [The photodynamic therapy (PDT) of the actinic keratoses, basal cell carcinomas and plantar warts]. derm - Praktische Dermatologie. 2005;11(4):301–304

Dickreiter B. Phototherapie - Therapeutische Möglichkeiten von Infrarotstrahlung und sichtbarem Licht [Phototherapy - therapeutic possibilities of infrared radiation and visible light]. Gesundes Leben. 2002;79(6):52–57

Meffert H, Müller GM, Scherf HP. Milde Infrarot-A-Hyperthermie zur Behandlung von Erkrankungen des rheumatischen Formenkreises. Anhaltende Verminderung der Aktivität polymorphkerniger Granulozyten [Mild infrared-A-hyperthermia for the treatment of diseases of the rheumatic disorders circle. Persistent decrease of the activity of granulocytes with polymorph nuclei]. Intern Sauna-Arch. 1993;10:125–129

Falkenbach A, Dorigoni H, Werny F, Gütl S. Wassergefilterte Infrarot-A-Bestrahlung bei Morbus Bechterew und degenerativen Wirbelsäulenveränderungen: Effekte auf Beweglichkeit und Druckschmerzhaftigkeit [Water-filtered infrared-A irradiation in Morbus Bechterew and degenerative vertebral column diseases: effects on flexibility and feeling of pressure]. Österr Z Physikal Med Rehab. 1996;6(3):96–102

Hoffmann G. Improvement of regeneration by local hyperthermia induced by waterfiltered infrared A (wIRA). Int J Sports Med. 2002;23(Suppl 2):S145

Singer D, Schröder M, Harms K. Vorteile der wassergefilterten gegenüber herkömmlicher Infrarot-Strahlung in der Neonatologie [Advantages of water filtered over conventional infrared irradiation in neonatology]. Z Geburtshilfe Neonatol. 2000;204(3):85–92. [PubMed]

Rowe E, Vinogradova I, Meffert H. Neue Methoden gegen Warzen: wIRA - effektiv und wirtschaftlich interessant [New methods against warts: wIRA - effective and commercially interesting]. Dtsch Dermatologe. 2004;52(7):487–489

Infrarot A läßt Warzen schmerzfrei verschwinden [Infrared-A causes warts to vanish without pain]. IGeL Plus. 2005;(1):54–55. . Available from: http://www.aerzte-zeitung.de/service/abo/igelbestellung/01IG_54-55.pdf.

Voss P, Hajimiragha H, Engels M, Ruhwiedel C, Calles C, Schroeder P, Grune T. Irradiation of GAPDH: a model for environmentally induced protein damage. Biol Chem. 2007;388(6):583–592. [PubMed]

Schieke S, Stege H, Kürten V, Grether-Beck S, Sies H, Krutmann J. Infrared-A radiation-induced matrix metalloproteinase 1 expression is mediated through extracellular signal-regulated kinase 1/2 activation in human dermal fibroblasts. J Invest Dermatol. 2002;119(6):1323–1329. [PubMed]

Schieke SM, Schroeder P, Krutmann J. Cutaneous effects of infrared radiation: from clinical observations to molecular response mechanisms. Photodermatol Photoimmunol Photomed. 2003;19(5):228–234. [PubMed]

Merk HF, Baron JM, Heise R, Fritsche E, Schroeder P, Abel J, Krutmann J. Concepts in molecular dermatotoxicology. Exp Dermatol. 2006;15(5):692–704. [PubMed]

Schroeder P, Pohl C, Calles C, Marks C, Wild S, Krutmann J. Cellular response to infrared radiation involves retrograde mitochondrial signaling. Free Radic Biol Med. 2007;43(1):128–135. [PubMed]

Kim MS, Kim YK, Cho KH, Chung JH. Regulation of type I procollagen and MMP-1 expression after single or repeated exposure to infrared radiation in human skin. Mech Ageing Dev. 2006;127(12):875–882. [PubMed]

Li WH, Lee YM, Kim JY, Kang S, Kim S, Kim KH, Park CH, Chung JH. Transient receptor potential vanilloid-1 mediates heat-shock-induced matrix metalloproteinase-1 expression in human epidermal keratinocytes. J Invest Dermatol. 2007;127(10:2328-35. DOI):10.1038/sj.jid.5700880 . Available from: http://dx.doi.org/10.1038/sj.jid.5700880.

Hartel M, Illing P, Mercer JB, Lademann J, Daeschlein G, Hoffmann G. Therapy of acute wounds with water-filtered infrared-A (wIRA) [review]. GMS Krankenhaushyg Interdiszip. 2007;2(2):Doc53 . Available from: http://www.egms.de/en/journals/dgkh/2007-2/dgkh000086.shtml.

Meffert H, author. Mögliche neue Indikationen - wIRA und kurzwelliges Licht zur Akne-Behandlung. Lecture presented at the course entitled "Water-filtered infrared-A in dermatology" of the Department of Dermatology of the University Hospital Jena in Jena, Germany, November 15, 2003 [Possible new indications - wIRA and short-wave light for the treatment of acne]. 2003

Aschoff R, author. Therapie der Akne papulopustulosa mittels vollem Spektrum von sichtbarem Licht und wassergefiltertem Infrarot A und/oder lokalem Aknetherapeutikum. Lecture presented at the symposium entitled "Water-filtered infrared-A (wIRA) in dermatology" of the Dr. med. h.c. Erwin Braun Foundation in Liestal/Basel, Switzerland, November 20, 2004 [Therapy of acne papulopustulosa with full spectrum of visible light and water-filtered infrared-A and/or topical acne therapeutic]. 2004

Huyke C, Schempp CM, authors. Klinische Anwendung von wIRA in der Universitätshautklinik Freiburg - Photodynamische Therapie PDT. Lecture presented at the symposium entitled "Water-filtered infrared-A (wIRA) in dermatology" of the Dr. med. h.c. Erwin Braun Foundation in Liestal/Basel, Switzerland, November 20, 2004 [Clinical applications of wIRA in the Department of Dermatology of the University Hospital Freiburg - Photodynamic Therapy PDT]. 2004

Villiger B, Fisch I, authors. Experiences with wIRA and movement in patients with fibromyalgia in Bad Ragaz, Switzerland. 2003-2005

Sattler H, Stellmann A, authors. Experiences with water-filtered infrared-A (wIRA) in patients during rehabilitation after hip and knee endoprosthesis operations in Bad Dürkheim, Germany. 2002-2004

von Lieven H, Dapper FD, authors. Water-filtered infrared-A radiation: a new technique for hyperthermia of superficial tumors in men. In: 1994. 14th Annual Meeting European Society Hyperthermic Oncology (ESHO); Amsterdam/Niederlande: p. P9

Hegewisch-Becker S, Jäger E, Knuth A, Riess H, Wiedemann GJ, Hossfeld DK. Ganzkörper-Hyperthermie in Kombination mit Chemotherapie. Deutsches Ärzteblatt. 2000;97:A2941–A2946

Carter DR, editor. Electro-Optics Handbook, Burle Industries, Lancester, PA, USA [reprint]. 1992. Baesweiler, Germany: Burle Industries GmbH; p. 63

Valley SL, author. Handbook of geophysics and space environments. 1965. Air Force Cambridge Research Laboratories, Office of Aerospace Research, U.S. Air Force;

[Back]

Figure 1. Spectral solar irradiance outside the atmosphere and at sea level

in both cases with the sun at the zenith and for a mean Earth-sun separation. Shaded areas indicate absorption at sea level due to the atmospheric constituents shown (from [97], adapted from [98]).

For comparison of Figures 1, 3 and 4: 1000 W m⁻² µm⁻¹ = 100 mW cm⁻² µm⁻¹ = 1 mW cm⁻² (10 nm)⁻¹

[Back]

[Back]

Figure 2. Cross-section of a water-filtered infrared-A radiator

(Hydrosun, Müllheim, Germany)

The whole incoherent broad-band radiation of a 3000 Kelvin halogen bulb is passed through a cuvette, containing water, which absorbs or decreases the undesired wavelengths within infrared (most parts of infrared-B and -C and the absorption bands of water within infrared-A). The water is hermetically sealed within the cuvette. A fan provides air cooling of the cuvette to prevent the water from boiling.

[Back]

[Back]

Figure 3. Spectral irradiance of a water-filtered infrared-A radiator

(Hydrosun® radiator 501 with 10 mm water cuvette and orange filter OG590) at approximately 210 mW/cm² (= 2.1 x 10³ W/m²) total irradiance (from [1]); (visible light (VIS): 380-780 nm; infrared-A (IR-A): 780-1400 nm; infrared-B (IR-B): 1400-3000 nm)

[Back]

[Back]

Figure 4. Comparison of the spectra of the sun at sea level and of a water-filtered infrared-A radiator

Spectral solar irradiance at sea level (with the sun at the zenith and for a mean Earth-sun separation) as in Fig. 1 (adapted from [97]) and spectral irradiance of a water-filtered infrared-A radiator (Hydrosun® radiator 501 with 10 mm water cuvette and orange filter OG590) at approximately 210 mW/cm² (= 2.1 x 10³ W/m²) total irradiance as in Fig. 3 (from [1]).

The spectrum of the sun at sea level includes ultraviolet radiation (UV, <400 nm), visible light (VIS, 380-780 nm), and infrared radiation (IR, >780 nm). The spectrum of the water-filtered infrared-A radiator includes only visible light (VIS) and infrared radiation (IR); the visible part depends on the used color filter; the wIRA radiator does not emit ultraviolet radiation (UV).

Both spectra show the decreased irradiance of the absorption bands of water.

[Back]

[Back]

Figure 5. Comparison of irradiation with water-filtered infrared-A and with conventional infrared

Thermographical comparison of skin surface temperatures in the lumbar region 12 minutes after beginning of irradiation with water-filtered infrared-A (left) and conventional infrared (right) with the same irradiance: the skin surface temperature is higher in case of irradiation with conventional infrared (presented in the thermography), while temperature in 1 cm depth of tissue is higher when irradiating with water-filtered infrared-A (from [45]). So water-filtered infrared-A presents a high tissue penetration combined with a low thermal load to the skin surface.

[Back]

[Back]

Figure 6. Example for an irradiation of a wound with a water-filtered infrared-A radiator

(published with kind approval of Prof. James Mercer, Tromsø/Norway) [27]

[Back]